• date post

22-Dec-2015
• Category

## Documents

• view

212
• download

0

Embed Size (px)

### Transcript of Homework # 1 2004200456 Lim Myeong-Jun. Index (Axid) Stress δ & ε Curve Cantilever Strain gage...

• Slide 1
• Homework # 1 2004200456 Lim Myeong-Jun
• Slide 2
• Index (Axid) Stress & Curve Cantilever Strain gage Gage factor Problem solution
• Slide 3
• (Axid) Stress M x y A F = M * g A = x * y On the surface, the average force per unit area is denoted as . stress cf) Pressure F is sometimes called Load : Strain(unitless) L F F
• Slide 4
• & Curve Brittle material ( ex. Glass) Linear : elastic region => Not linear over a range n : Ultimate stress
• Slide 5
• & Curve Ductile material ( ex. Al, Steel) n : Ultimate Stress y : yield Stress pl : Proportional Limit
• Slide 6
• & Curve For elastic region (linear region) ( pl) = E * E : Youngs modulus ( Modulus of elastivity)
• Slide 7
• Cantilever L F F E : constant, A & L : almost constant If we know , then we can find F (force).
• Slide 8
• Strain gage A strain gauge (alternatively: strain gage) is a device used to measure the strain of an object. Invented by Edward E. Simmons and Arthur C. Ruge in 1938, the most common type of strain gauge consists of an insulating flexible backing which supports a metallic foil pattern. The gauge is attached to the object by a suitable adhesive, such as cyanoacrylate. As the object is deformed, the foil is deformed, causing its electrical resistance to change. This resistance change, usually measured using a Wheatstone bridge, is related to the strain by the quantity known as the gauge factor.
• Slide 9
• Strain gage The black line is electrical wire. A L +V-+V- The equation shows relation between strain and resistance. If we push the strain, , A and L was changed.
• Slide 10
• Strain gage when you differentiate each variable, The third equations - sign means that opposite change as compared with change of resistance. -> R L, R 1/A Total change of resistance
• Slide 11
• Strain gage Where : Poissons ratio D : D/D = strain along the diametrical(horizontal) axis. L : L/L = strain along the longitudinal(vertical) axis.
• Slide 12
• Strain gage & Gage factor Piezoresistive effect Dimensional effect Gage factor For mental strain gage : G = ~1.6 For semi conductor strain gage : G = 100 ~ 170 high temperature coefficient
• Slide 13
• Gage factor The gauge factor GF is defined as: where R G is the resistance of the undeformed gauge, R is the change in resistance caused by strain, and is strain. For metallic foil gauges, the gauge factor is usually a little over 2. For a single active gauge and three dummy resistors, the output v from the bridge is: where BV is the bridge excitation voltage. Foil gauges typically have active areas of about 2-10 mm 2 in size. With careful installation, the correct gauge, and the correct adhesive, strains up to at least 10% can be measured.
• Slide 14
• Problem # 3 Gage factor : 10 Gage 1 & 2 : L -> L + L Gage 3 & 4 : L -> L - L F Side view Top view 1 2 3 4 1 2 L F
• Slide 15
• Problem # 3 V0 = Av(Va Vb) A : Differential-mode voltage gain Cf) : Resistor : variable Resistor : Resistive Sensor IA -+-+ Vb Va
• Slide 16
• Problem # 3 VaVb
• Slide 17
• Problem # 3 - The constant 50 defined by gage factor(10) and voltage source(5v). - Av is changeable factor by resistor connected IA(Instrument Amplifier) - k is given by structure and material. - There are linearity between Vo and F.
• Slide 18
• Problem # 4 Gage factor : p-type Si strain gage S1 & S2 : G = 100 n-type Si strain gage S3 & S4 : G = -100 1 3 F Side view Top view 1 3 2 4 1 3
• Slide 19
• Problem # 4 E
• Slide 20
• Slide 21
• F Side view 1 3 2 4 Use least squares method to find the calibration equation. Force range : 0~100N
• Slide 22
• Problem # 6 In the below pressure sensor, diaphragm is equipped with two p- type Si strain gage of gage factor of +100 and two n-type Si strain gage of gage factor of -100. As pressure is placed in the diaphragm, the same strain occurs in the each strain gage and the responsiveness is 10 -5 %/mmHg. As not, the strain gage is 50. suppose the relationship between pressure and strain is linear. (a) When pressure is changed from 0 to 50mmHg, What is the resistance of the p-type and Si strain gage changed? (b) Sketch the bridge circuit consisting of four strain gage and mark where each strain gage is placed. (c) Let the driving power source have voltage of 1 and add instrumentation amplifier for the final output to be converted from 0 to 1V. What is the gain?
• Slide 23
• Problem # 6 Gage factor : p-type Si strain gage S1 & S2 : G = 100 n-type Si strain gage S3 & S4 : G = -100 2424 1313
• Slide 24
• Problem # 6 (mmHg) () p-type n-type 50 0 500 50.25 49.75
• Slide 25
• Problem # 6 1
• Slide 26
• Slide 27
• Thanks for your attention!!